<?xml version='1.0' encoding='ISO-8859-1'?>
<!-- 

-->
<!-- *DewarPort="10000" -->
<!-- LNAPort="10001" -->

<!-- timeout in microseconds -->
<MedicinaKBandDualFReceiver
    xmlns="urn:schemas-cosylab-com:MedicinaKBandDualFReceiver:1.0" 
    xmlns:baci="urn:schemas-cosylab-com:BACI:1.0" 
    xmlns:cdb="urn:schemas-cosylab-com:CDB:1.0" 
    xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance" 
    actionThreadStackSize="2048"
    monitoringThreadStackSize="4096"
    DewarIPAddress="192.168.51.2"
    DewarPort="10000"
    LNAIPAddress="192.168.51.2"
    LNAPort="10001"   
    VertexIPAddress="192.168.51.2"
    VertexPort="2096"
    VertexCommand="proc kkc\n"   
    WatchDogResponseTime="10000000"
    WatchDogSleepTime="10000000"
    LNASamplingTime="25000000"
    RepetitionCacheTime="7000000"
    RepetitionExpireTime="10000000"
    LocalOscillatorInstance="RECEIVERS/LO"
    ReceiverName="KBandDualFReceiver"
>
    <LO description="Sequence of local oscillator values for each IF chain" />
    <feeds description="Number of feeds of the receiver" />
    <IFs description="Number of Intermediate Frequencies chains coming from each feed" />
    <polarization description="Sequence of values that identifies the polarization of each IF chains" />    
    <initialFrequency description="Sequence of the start frequencies for each IF chain (MHz)" 
                      units="MHz"
                      format="%6.2f"
    />  
    <bandWidth description="Sequence of the band width values for each IF chain (MHz)" 
               units="MHz"
               format="%6.2f"
    />
    <mode description="Mnemonic code that defines a special working mode of the receiver (MHz)" />
    <vacuum description="measure of the vacuum"
            alarm_high_on="5e-3"
            default_value="5e-7"
            alarm_high_off="5e-6"
            alarm_timer_trig="0"
            alarm_fault_family="BACIProperty"
            alarm_fault_member="KBandDualFReceiver"
            alarm_level="0"
            default_timer_trig="10"
            min_timer_trig="1"
            min_delta_trig="1e-7"
            units="mBar"
            format="%e"
    />
    <status description="Bit fields that provide the status word of the receiver and component"
            alarm_mask="777"
            alarm_trigger="1"
            alarm_timer_trig="1"
            alarm_fault_family="BACIProperty"
            alarm_fault_member="KBandDualFReceiver"
            alarm_level="0"
            default_timer_trig="10"
            min_timer_trig="1"
    />  
    <vdL1 description="drain voltage for the LNA left channel, stage 1"
          alarm_high_on="1.41"
          alarm_low_on="1.01"
          default_value="1.21"
          alarm_high_off="1.31"
          alarm_low_off="1.11"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />
    <vdR1 description="drain voltage for the LNA right channel, stage 1"
          alarm_high_on="1.71"
          alarm_low_on="0.91"
          default_value="1.31"
          alarm_high_off="1.61"
          alarm_low_off="1.01"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />
    <vdL2 description="drain voltage for the LNA left channel, stage 2"
          alarm_high_on="1.40"
          alarm_low_on="1.00"
          default_value="1.20"
          alarm_high_off="1.30"
          alarm_low_off="1.10"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <vdR2 description="drain voltage for the LNA right channel, stage 2"
          alarm_high_on="1.70"
          alarm_low_on="0.9"
          default_value="1.3"
          alarm_high_off="1.60"
          alarm_low_off="1.00"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <vdL3 description="drain voltage for the LNA left channel, stage 3"
          alarm_high_on="1.70"
          alarm_low_on="1.30"
          default_value="1.50"
          alarm_high_off="1.60"
          alarm_low_off="1.40"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <vdR3 description="drain voltage for the LNA right channel, stage 3"
          alarm_high_on="1.95"
          alarm_low_on="1.05"
          default_value="1.45"
          alarm_high_off="1.85"
          alarm_low_off="1.15"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <vdL4 description="drain voltage for the LNA left channel, stage 4"
          alarm_high_on="1.72"
          alarm_low_on="1.32"
          default_value="1.52"
          alarm_high_off="1.62"
          alarm_low_off="1.42"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <vdR4 description="drain voltage for the LNA right channel, stage 4"
          alarm_high_on="1.75"
          alarm_low_on="1.15"
          default_value="1.45"
          alarm_high_off="1.65"
          alarm_low_off="1.25"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <vdL5 description="drain voltage for the LNA left channel, stage 5"
          alarm_high_on="1.71"
          alarm_low_on="1.31"
          default_value="1.51"
          alarm_high_off="1.61"
          alarm_low_off="1.41"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <vdR5 description="drain voltage for the LNA right channel, stage 5"
          alarm_high_on="1.77"
          alarm_low_on="1.17"
          default_value="1.47"
          alarm_high_off="1.67"
          alarm_low_off="1.27"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <idL1 description="drain current for the LNA left channel, stage 1"
          alarm_high_on="2.83"
          alarm_low_on="2.03"
          default_value="2.43"
          alarm_high_off="2.73"
          alarm_low_off="2.13"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <idR1 description="drain current for the LNA right channel, stage 1"
          alarm_high_on="17.00"
          alarm_low_on="0.5"
          default_value="7.50"
          alarm_high_off="15.00"
          alarm_low_off="1.5"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <idL2 description="drain current for the LNA left channel, stage 2"
          alarm_high_on="3.0"
          alarm_low_on="2.3"
          default_value="2.7"
          alarm_high_off="2.9"
          alarm_low_off="2.4"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <idR2 description="drain current for the LNA right channel, stage 2"
         alarm_high_on="17.00"
          alarm_low_on="0.5"
          default_value="7.50"
          alarm_high_off="15.00"
          alarm_low_off="1.5"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <idL3 description="drain current for the LNA left channel, stage 3"
          alarm_high_on="3.60"
          alarm_low_on="2.60"
          default_value="3.10"
          alarm_high_off="3.40"
          alarm_low_off="3.20"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <idR3 description="drain current for the LNA right channel, stage 3"
          alarm_high_on="17.00"
          alarm_low_on="0.5"
          default_value="7.50"
          alarm_high_off="15.00"
          alarm_low_off="1.5"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <idL4 description="drain current for the LNA left channel, stage 4"
          alarm_high_on="3.57"
          alarm_low_on="2.57"
          default_value="3.07"
          alarm_high_off="3.37"
          alarm_low_off="2.87"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <idR4 description="drain current for the LNA right channel, stage 4"
          alarm_high_on="5.00"
          alarm_low_on="0.0"
          default_value="1.50"
          alarm_high_off="4.00"
          alarm_low_off="0.02"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <idL5 description="drain current for the LNA left channel, stage 5"
          alarm_high_on="0.2"
          alarm_low_on="-0.22"
          default_value="-0.007"
          alarm_high_off="0.1"
          alarm_low_off="-0.1"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <idR5 description="drain current for the LNA right channel, stage 5"
          alarm_high_on="0.3"
          alarm_low_on="-0.1"
          default_value="0.02"
          alarm_high_off="0.2"
          alarm_low_off="-0.05"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <vgL1 description="gate voltage for the LNA left channel, stage 1"
          alarm_high_on="0.1"
          alarm_low_on="-0.045"
          default_value="-0.10"
          alarm_high_off="0.05"
          alarm_low_off="-0.03"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <vgR1 description="gate voltage for the LNA right channel, stage 1"
          alarm_high_on="0.066"
          alarm_low_on="-0.045"
          default_value="-0.08"
          alarm_high_off="0.05"
          alarm_low_off="-0.03"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <vgL2 description="gate voltage for the LNA left channel, stage 2"
          alarm_high_on="0.05"
          alarm_low_on="-0.2"
          default_value="-0.09"
          alarm_high_off="0.048"
          alarm_low_off="-0.15"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <vgR2 description="gate voltage for the LNA right channel, stage 2"
          alarm_high_on="0.059"
          alarm_low_on="-0.30"
          default_value="-0.10"
          alarm_high_off="0.048"
          alarm_low_off="-0.2"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <vgL3 description="gate voltage for the LNA left channel, stage 3"
          alarm_high_on="0.052"
          alarm_low_on="-0.060"
          default_value="-0.14"
          alarm_high_off="0.4"
          alarm_low_off="-0.04"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <vgR3 description="gate voltage for the LNA right channel, stage 3"
          alarm_high_on="0.052"
          alarm_low_on="-0.26"
          default_value="-0.10"
          alarm_high_off="0.04"
          alarm_low_off="-0.2"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />


    <vgL4 description="gate voltage for the LNA left channel, stage 4"
          alarm_high_on="0.049"
          alarm_low_on="-0.179"
          default_value="-0.12"
          alarm_high_off="0.03"
          alarm_low_off="-0.158"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />
    <vgR4 description="gate voltage for the LNA right channel, stage 4"
          alarm_high_on="5.0"
          alarm_low_on="-0.4"
          default_value="0.0"
          alarm_high_off="4.8"
          alarm_low_off="-0.3"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />
    <vgL5 description="gate voltage for the LNA left channel, stage 5"
          alarm_high_on="0.089"
          alarm_low_on="-0.075"
          default_value="0.0"
          alarm_high_off="0.088"
          alarm_low_off="-0.074"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />
    <vgR5 description="gate voltage for the LNA right channel, stage 5"
          alarm_high_on="4.4"
          alarm_low_on="3.4"
          default_value="3.9"
          alarm_high_off="4.3"
          alarm_low_off="3.5"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.05"
          units="Volts"
          format="%5.2f"
    />

    <cryoTemperatureCoolHead description="Cryogenic temperature measured near the cool head"
          alarm_high_on="10.0"
          alarm_high_off="-10.0"
          alarm_low_on="5.0"
          alarm_low_off="-5.0"
          default_value="0.0"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.5"
          units="Kelvin"
          format="%5.1f"                                    
    />
    <cryoTemperatureCoolHeadWindow description="Cryogenic temperature measured at cool head window"
          alarm_high_on="10.0"
          alarm_high_off="5.0"
          alarm_low_on="-10.0"
          alarm_low_off="-5.0"
          default_value="0.0"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.5"
          units="Kelvin"
          format="%5.1f"                        
    />
    <cryoTemperatureLNA description="Cryogenic temperature measured at the LNA"
          alarm_high_on="30.0"
          alarm_high_off="27.0"
          alarm_low_on="4.0"
          alarm_low_off="8.0"
          default_value="20.0"
          alarm_timer_trig="1.0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="1.0"
          min_timer_trig="1"
          min_delta_trig="0.5"
          units="Kelvin"
          format="%5.1f"                                    
    />
    <cryoTemperatureLNAWindow description="Cryogenic temperature measured at the LNA window"
          alarm_high_on="85.0"
          alarm_high_off="82.0"
          alarm_low_on="44.0"
          alarm_low_off="54.0"
          default_value="70.0"
          alarm_timer_trig="0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.5"
          units="Kelvin"
          format="%5.1f"                                                
    />
    <environmentTemperature description="temperature of the room in which the receiver is installed"
          alarm_high_on="38.0"
          alarm_high_off="36.0"
          alarm_low_on="15.0"
          alarm_low_off="17.0"
          alarm_fault_family="BACIProperty"
          alarm_fault_member="KBandDualFReceiver"
          default_value="24.0"
          alarm_timer_trig="1.0"
          alarm_level="0"
          default_timer_trig="10"
          min_timer_trig="1"
          min_delta_trig="0.5"
          units="Celsius degree"
          format="%4.1f"                                    
    />
    <receiverStatus description="Overall status of the receiver"/>

    <receiverName description="ReceiverName"/>

</MedicinaKBandDualFReceiver>
